I hope you are doing well
when I made an Autotune on GC-MS to check the apparatus, it does not continue the process, and there a problem in the mass Filter Electronics as shown in the pictures.
If any suggestion for the proper solution for that problem? So please share your thoughts.
Thanks for your kind attention.
Hiiiii everyone! I have an enquiry on statistical analysis. I was looking for many forum and it's still cannot solve my problem. I want to compare means of two groups of data but only with two...
03 March 2021 8,796 3 View
I would like to create a small guide containing the meanings of the names .. Thanks to everyone who cooperates with me in achieving it.
03 March 2021 5,266 4 View
Is There Any Feasible Method To Test The Efficiency Of Fluorescent Compounds Other Than UV Spectrometers ? Suggestions Would Be Appreciated !
02 March 2021 5,785 3 View
How to do Piezoelectric coupling in ANSYS-APDL or COMSOL between tapered substrate and rectangular piezoelectric patch? Piezoelectric coupling between the rectangular substrate and rectangular...
02 March 2021 9,833 3 View
I am going to have 3 different probes in my qPCR work that I am going to do. But I realized that the machine we have in the lab is a Rotor-Gene Q 2plex HRM Platform, saying it has green, yellow,...
01 March 2021 8,544 1 View
Is the electropolymerisation of polyaniline coating over stainless steel (SS) via Cyclic voltammetry technique is really adherant? Else how to improve the adhesion of the coating, whether any...
01 March 2021 8,220 2 View
I have created an Ice 1h crystalline structure and an Aluminium substrate structure and equilibrated both at 250K. Now I need to bring them together in a way that an adhesive bond is created....
01 March 2021 3,325 2 View
Does anyone have the experience of using Taq Man probes in the QIAGEN Rotar- Gene qPCR machine?
01 March 2021 5,311 1 View
When explaining substitution models, the substitutions are expressed as Q matrix. Why is the sum of the elements in a row zero?
28 February 2021 3,864 3 View
Hi, I implemented a code to gabor filter cifar10 data but the images after being filtered and stacked are not clear like the original images. I think the problem is in the way I am using the...
03 March 2021 6,317 1 View
Hi, I am after the reference below, my library says it cannot obtain a copy either locally or internationally, any help appreciated! Chris Wang ZM, Heshka S, Wielopolski L, Pi-Sunyer FX, Pierson...
03 March 2021 6,193 1 View
Hi, I hope that someone can help me with this task: I have 5 groups of participants which are numbered in SPSS from 1 to 5 (according to their group number) and are all under the variable group....
03 March 2021 506 3 View
I am using a 2707 waters HPLC device. When I try to inject a sample, it says missing plate or rack. I changed the needle and calibrated its position but I still get the same problem. I even get...
02 March 2021 1,408 1 View
I feel that the practice in teacher education in my country is below the expected performance level due to very poor management system. Hope I will learn something from your experiences.
02 March 2021 1,516 4 View
Hello! I'll do a size exclusion chromatography, but I only have an open column, and I'll perform the peptide extraction from yeast, using buffer lysis (sodium phosphate 50 mM/NaCl 30 mM/DNAse and...
01 March 2021 2,215 2 View
I analyzed flavonoid total spectrophotometry, the procedure as below: 2 ml supernatan added 5 ml dd.H2O, then added 0,15 ml NaNO2 5%, 5 minutes later added 0,15 ml AlCl3 10%, After 6 minutes,...
01 March 2021 6,722 1 View
Hi, I am running a size exclusion chromatography experiment with a buffer containing Potassium Acetate as a salt. I analyse these fractions through SDS-PAGE. After boiling my SEC fractions in...
01 March 2021 2,622 3 View
I am doing my thesis on the nutrient deprivation on the macroalgae Rhizoclonium sp. I will be analysing the levels over a period of roughly 5 days by HPLC-UV and GC/MS. I will be using seawater as...
28 February 2021 6,477 1 View
In a such case, we usually call service man. because this isn't an issue like leak that we can solve with some tries.
But If you are an expert and can open the mass unit, this error would be solved with cleaning.
You should remove quadrupole contacts firstly and clean them with very pure IPA or MeOH. Thereafter reinstall all of them based on the instrument guideline. you must be then ensured from accurate placement of all connections and the entire wires very carefully.
That is a general purpose latchup error. Typically it indicates that you have an electronic error, but there are other error states that can give you that, such as insufficient vacuum. Check your vacuum status first, then check your electronic status (MS Check). You may be able to change your scan range and get the system to partially work. When was the last time you cleaned the system and serviced the pumps?
In my opinion, one of the reason of Your troubles may be contaminated the ion source and quadrupole systems. Please, made carefully the cleaning operation of these systems. If it, doesn't help - call the professional service.